Ø8 mm (0.315 in.)
8.9 mm (0.35 in.) Max.
(10X ns, 10 fs)
9.5 mm (0.37 in.) Max.
…Materials Characterization Parameter Value Laser Wavelength 785 nm Laser Power Up to 400 mW at sample Spectral Range -800 cm -1 to 2800 cm -1 Spectral Dispersion 2 cm -1 per pixel Focusing Manual or Auto Objectives 4 user definable (e.g. 5x, 10x, 20x) Maximum Sample Size 125 mm x 85 mm (fits…
round laser module (loosing up and setting the two set screws). The modular beam
expander and spatial filter from Newport was tested with the laser module (figure 5.).
Figure 4:free space module secured in a threaded adapter
A combination of microscope objective 20X and 10…
Class 1 Housing
20x Microscope Objective
Manual Mask Slider Assembly
Mask "Small Spots"
Mask "Square and Round Spots"
Mass Flow Controller
Sample Chamber Window 2'
MMG Sample Cell
GeoLasHD Excimer Lasers and UV Optical Systems
Up to 400mW at sample
-800cm-1 to 2800cm-1
2 cm-1 per pixel
Manual or Auto
4 user definable (e.g. 5x, 10x, 20x)
125mm x 85mm (fits standard well plate)
110-240 VAC ±…
Please Read Before Continuing
Risk factors: Except for the historical information contained here, many of the matters discussed in this Web site are forward-looking statements, based on expectations at the time they were made, that involve risks and uncertainties that could cause our results to differ materially from those expressed or implied by such statements. These risks are detailed in the “Factors That May Affect Future Results” section of our latest 10-K or 10-Q filing. Coherent assumes no obligation to update these forward-looking statements.