…protective cover, not the LDFP (Low Distortion
Faceplate). See the following figure.
Signal Words in this Document
This document uses the NOTICE signal word.
Indicates information considered important,
but not hazard-related.
…13 mW/cm2 (with LDFP) at 633 nm 5 mW/cm2 (with LDFP) at 633 nm
5 µW/cm2 (without LDFP) at 633 nm 2 µW/cm2 (without LDFP) at 633 nm
70 mW/cm2 (with LDFP) at 1064 nm 25 mW/cm2 (with LDFP) at 1064 nm
340 µW/cm2 (without LDFP) at 1064 nm 125 µW/cm2 (without LDFP) at 1064 nm
® ® 50.0 50.0 50.0 64.3 68.3 59.9 17.5 21.5 IMAGE PLANE PROTECTIVE CAP LDFP ADAPTER M6 THD 2X 25.0 1/4-20 UNC-2B THD 2X 46.2 SCALE = 1:1 SIZE WT: DWG. NO. B SHEET 1 OF 1 REV DATE APPROVALS DFT: J. WILSON CHK: ENG: APR: MATERIAL: FINISH: NEXT ASSY PROPRIETARY NOTICE: THE INFORMATION CONTAINED IN THIS…
…with LDFP 13 mW/cm2 at 633 nm 5 mW/cm2 at 633 nm 90 mW/cm2 at 248 nm5
without LDFP 5 W/cm2 at 633 nm 2 W/cm2 at 633 nm 90 µW/cm2 at 248 nm4
with LDFP 70 mW/cm2 at 1064 nm 25 mW/cm2 at 1064 nm
without LDFP 340 W/cm2 at 1064 nm 125 W/cm2 at 1064 nm
…In (TTL) Saturation CW (at 1064 nm) 3.5 mW/cm2 (with LDFP), 50 µW/cm2 (without LDFP) Saturation CW (at 1523 nm) 350 µW/cm2 (with LDFP), 30 µW/cm2 (without LDFP) Saturation Pulse (at 1064 nm) 5 µJ/cm2 (with LDFP), 0.08 µJ/cm2 (without LDFP) Sensor Active Area (mm)(H xW) 9.6 x 7.7 Sensor Elements…
…Low-Distortion Face Plate (LDFP) and each BCUBE as one optic, and the VARM or C-VARM as three optics each. Attenuator Selection Attenuation is selected on the basis of power density in W/cm2 or energy density in J/cm 2. The attenuation from the camera’s Low-Distortion Face Plate (LDFP) will allow an…
* Ease-of-use connectivity
* USB bus-powered low voltage operation
* Broad spectral range:
* 190 nm to 1100 nm
* 400 nm to 1100 nm (with LDFP)
* 190 nm to 355 nm (with BIP-12F)
* 190 nm to 355 nm (UV model)
* Large dynamic range
* Digital output through USB 2.0 eliminates the need for an…
…Pulse Trigger in Rate3 (Hz) 200 200 200
without LDFP 32 mJ/cm2 at 1064 nm 32 mJ/cm2 at 1064 nm 200 µJ/cm2 at 1064 nm4
13 mW/cm2 at 633 nm
5 W/cm2 at 633 nm
70 mW/cm2 at 1064…
…camera, remove only the
protective cover, not the LDFP (Low Distortion Faceplate).
See the following figure.
Do not connect a LaserCam-HR camera to the USB 2.0 PC
port before starting the following installation.
Insert the BeamView-USB CD into…
® ® 68.1 68.1 79.3 11.2 Protective Cap LDFP Adapter Rotational Mount 74.3 18.7 Image Plane 41.8 27.2 1/4-20 UNC 9.9 Top View Side View Bottom View Front View SCALE = 1:1.25 SIZE WT: DWG. NO. B SHEET 1 OF 1 REV DATE APPROVALS DFT: J. WILSON CHK: ENG: APR: MATERIAL: FINISH: NEXT ASSY PROPRIETARY…
Please Read Before Continuing
Risk factors: Except for the historical information contained here, many of the matters discussed in this Web site are forward-looking statements, based on expectations at the time they were made, that involve risks and uncertainties that could cause our results to differ materially from those expressed or implied by such statements. These risks are detailed in the “Factors That May Affect Future Results” section of our latest 10-K or 10-Q filing. Coherent assumes no obligation to update these forward-looking statements.