…battery is drained. The meter can also operate with six standard AA batteries. Specifications Max. Measurable Power (with thermopile sensor) 30.0 kW Max. Measurable Power (with optical sensor) 300.0 mW Max. Measurable Energy (with pyroelectric sensor) 300.0J Min. Measurable Power (with thermopile…
…SECTION THREE: DESCRIPTION
Thank you for purchasing the FieldMaxII-TOPTM--a versatile,
easy-to-use digital power/energy meter designed for field service
and production applications.
In this section:
· Front panel (p. 3-2)
· Right side panel (p. 3-4)
· Left side…
…Several bug fixes including an auto-range timeout error that affected some customers using remote PC control Specifications Max. Measurable Power (with thermopile sensor) 30.0 kW Max. Measurable Power (with optical sensor) 300.0 mW Min. Measurable Power (with thermopile sensor) 10 µW Min. Measurable…
…batteries Specifications Max. Measurable Energy (with pyroelectric sensor) 300.0J Min. Measurable Energy (with pyroelectric sensor) 1 nJ Displayable Full-Scale Energy Range (with pyroelectric sensor) 3.000 nJ to 300.0J (manual ranging) Measurement Resolution 0.1% of full scale Max. Rep. Rate (pulsed…
…UW-150 RT Rotary
Height: 76" Height: 70"
Width: 68" Width: 32"
Depth: 38.0" Depth: 60.0"
Work Envelope: XYZ, 30" x 30" x 19"
Max distance between base plate and
galvo box: 22"
Standard Features for all UW models:
! Manual Z axis, internal adjustment
…4 LASER PRODUCT PER EN
635 nm to 670 nm VISIBLE (1 mW)
9.1 µm to 10.8 µm INVISIBLE
2000W MAX. AVG. 6J MAX. 1 ms PULSE
Please Read Before Continuing
Risk factors: Except for the historical information contained here, many of the matters discussed in this Web site are forward-looking statements, based on expectations at the time they were made, that involve risks and uncertainties that could cause our results to differ materially from those expressed or implied by such statements. These risks are detailed in the “Factors That May Affect Future Results” section of our latest 10-K or 10-Q filing. Coherent assumes no obligation to update these forward-looking statements.